Phenomenon: the flow path is connected with the suppression of current, the waste tube is not a continuous bubble discharge.
Possible cause: 1 Supporter positive and negative power cord is not connected.
2 suppressor internal aging, inhibition capacity decline.
Remedy: 1 Connect the suppressor electrode correctly or increase the suppression current
2 It is recommended to replace the suppressor